In all screens however, the top and bottom lines always display the same information: the top line displays system information, and the bottom line always gives the softkey options. The following is an example of a typical resting ECG screen. When battery capacity is limited the battery symbol flashes. Pressing the key below the text carries out the function indicated.
|Published (Last):||6 November 2015|
|PDF File Size:||6.38 Mb|
|ePub File Size:||6.17 Mb|
|Price:||Free* [*Free Regsitration Required]|
In all screens however, the top and bottom lines always display the same information: the top line displays system information, and the bottom line always gives the softkey options. The following is an example of a typical resting ECG screen. When battery capacity is limited the battery symbol flashes. Pressing the key below the text carries out the function indicated. The options available will change according to the screen displayed.
Items 4 to 10 are specific for ECG acquisition only: 4 Current Heart Rate averaged over 4 beats and refreshed every 2 seconds. The HR is also given on a manual printout. Note that with an auto mode printout the HR is averaged over the full 10 seconds of the recording. The electrode s must be reapplied. Change the sensitivity with the keys 3 auto , 4, 5 and 6. An A in this box indicates that automatic sensitivity is selected auto mode printout only.
Change the speed with the keys 8 and 9. The filter is applied with the filter key. The frequency of the filter cutoff is defined on page 35 menu Filters. Do not operate the unit if the earth connection is suspect or if the mains lead is damaged or suspected of being damaged. Page 14 Art. Refer to chapter 6. Connect the power cable at the rear of the unit. The mains indicator lamp 6 is always lit when the unit is connected to the mains supply. If the unit is switched on, the relevant symbol is displayed on the LCD 7.
Leave the P connected to the mains for 7 hours to fully charge the battery. Connect the potential equalisation cable and all other necessary cables at the rear of the P The patient data or the ECG acquisition screen is displayed see paragraph 5. Check the settings according to chapters 5. Connect the patient cable on the right side panel. The power source is indicated on the top line of the LCD. The internal battery provides power for up to 3 hours. When the unit is running on battery power a battery symbol 1 is displayed.
When working from battery power, the unit is automatically switched off after 5 minutes 30 seconds if battery capacity is limited if no key is pressed. The unit can remain connected to the mains supply without damage to either the battery or the unit.
Use the hospital or building common ground To prevent the possibility of leakage current when an external printer is connected, always ensure that the mains lead, or the potential equalisation is attached to the P Art.
E.C.G. 6/12 CANALI ESAOTE INTERPRETATIVO MOD. P8000 POWER
Esaote P-8000 Power ECG / EKG Machine
Esaote P8000 - User Manual