All rights reserved. Printed in USA. Specifications are subject to change without notice. All product names are trademarks of their respective companies. Page 2 Fluke authorized resellers shall extend this warranty on new and unused products to end-user customers only but have no authority to extend a greater or different warranty on behalf of Fluke.
|Published (Last):||24 February 2008|
|PDF File Size:||12.25 Mb|
|ePub File Size:||20.50 Mb|
|Price:||Free* [*Free Regsitration Required]|
Check the test leads for continuity. Replace damaged test leads before using the Tester. July Rev. Printed in USA. All product names are trademarks of their respective companies. Protection provided by the Tester may be impaired. When in doubt, have the Tester serviced. Do not dispose of this product as unsorted municipal waste. Page 5: Performance Tests 3. Replace the battery door and secure it by turning the battery door fastener one- quarter turn clockwise. The beeper sounds unless disabled and the Tester switches to continuity mode.
To complete the self test, measure a known voltage before use. If the Tester fails self test, it must not be used and requires service. This will force a processor reset that will display all LCD segments for approximately 1 second at power up. Table 3. Page 11 Limited Warranty and Limitation of Liability Each Fluke product is warranted to be free from defects in material and workmanship under normal use and service.
The warranty period is two years and begins on the date of shipment. Parts, product repairs, and services are warranted for 90 days. This manual is also suitable for:.
FLUKE 1550C Insulation Resistance Tester (5kV) / 250V – 5000V
Fluke 1550C Insulation Resistance Tester 5 kV
Providing the finest test equipment solutions since 1992