Not all applications are simple, you know. High temperature, foam, tank design, and other things can create big challenges in level measurement. Some companies tackle these challenges with radar. Radar transmitters have gained ground in level applications, even nudging differential pressure transmitters from the top spot in some areas.
|Published (Last):||4 April 2010|
|PDF File Size:||20.86 Mb|
|ePub File Size:||7.21 Mb|
|Price:||Free* [*Free Regsitration Required]|
Faemi On the other hand, the beam angle gave me less trouble. You can set up the output through the local display or a handheld. However, some places resist change, even beneficial change. To show that radar can work in a tank with a stirrer if the orohne constant could support it, we tested both the wireless communication and the radar device over a week. Of course, your application may need advanced configuration and analysis. Well, it does a good job of measuring the difference between the emitted signal and the received signal.
Check it out and see if it suits your process. Most of the time, you install the radar on top of the vessel to avoid waiting for a scheduled downtime. Of course, the secondary values mean more to you, right? You can check the reflection graph to do a deep analysis to clear signal interference or measure tanks with curved or conical bottoms. On to the review. Both technologies passed with flying colors. Please check your inbox. Sie finden sie in unserem Multimarken- eShop.
Sind Sie bereits Mitglied? You can use it to measure ranges up to 80 meters. Diese Information hilft uns um eine optiwwve Community bei Visaya aufzubauen. We sent you a message with further instructions. I once worked on a level measurement project for a food company that wanted to test radar and wireless communication. Take a seat, grab a cola, have a read, and draw your conclusions! Some companies tackle these challenges with radar. Because it addresses the integration issue fairly well, and that issue alone may tip the scales between two vendors.
Would you like to sign in? These iptiwave protocols will probably fit your process easily. This video will tell you more: What can it do?
Empfohlene Artikel Product Review: Radar transmitters have gained ground in level applications, even nudging differential pressure transmitters from the top spot in some areas.
This email is already in use. Kourtesy of Krohne Most of the time, you install the radar on top of the vessel to avoid waiting for a scheduled downtime. Why should I care? High temperature, foam, tank design, and other things can create big challenges in level measurement. However, the customer had a bias against the radar because the rest of the plant ran with differential pressure transmitters.
Kourtesy of Krohne It also has a step-by-step guide for a quick setup. Courtesy of Giphy This experience shows that people avoid unfamiliar tech, even if it proves useful. Not all applications are simple, you know.
TOP Related Posts.
KROHNE Product Downloads – OPTIWAVE 6400
Willkommen bei KROHNE Deutschland
KROHNE Download merkezi – tüm ürünler için Manueller